The 9600FLT 1 GHz Wide Offset Phase Noise Filter accessory is for high-performance spectrum analyzer wide-offset phase noise tests.
Use the Filter only at 1 GHz. It rejects phase noise at offsets greater than 1 MHz from the 1 GHz carrier (both sidebands are rejected). Use the filter only when the phase noise specification of the 96xx RF Reference Source is not considered sufficient for the target workload. Only the highest performance spectrum analyzers are likely to require the filter. The filter connects in series with the umbilical lead of the 9640A-50 or 9640A-75 Leveling Head. Use the supplied mounting kit to conveniently attach the filter to the front left handle of the Reference Source. This mount is fully compatible with the Y9600 Rack Mount accessory.
|Dimensions||130 mm (5.1 in) wide, 84 mm (3.3 in) high and 18 mm (0.7 in) deep.|
|Weight||0.25 kg (0.6 lbs)|
|Connectors||Input and Output: SMA female|
|Mounting ||Suitable for self-supporting use or optional mounting to the model 9640A RF Reference
Source left-side handle in bench top or rack mounted applications.
| The Filter is inserted between the 9640A front panel RF output and leveling head cable connectors, only when required for phase noise tests at 1 GHz output frequency. The leveling head cable connects directly to the front panel for all other applications.|
|Filter Type||Band-pass, narrow bandwidth|
|Offset Frequency (from 1 GHz)||≤300 kHz||1 MHz||3 MHz||10 MHz|
|Insertion Loss||2 dB||2.5 dB||5 dB||17 dB|
|Rejection||0 dB||0.5 dB||3 dB||15 dB|
| Attenuation of phase noise sidebands relative to the carrier level when used to reduce wide offset phase noise of a 1 GHz signal source.|