For secondary level performance with full ITS-90 calibration, Fluke Calibration’s 5618B series PRTs are an excellent choice for temperature measurements
With each probe you will receive a full NVLAP-accredited calibration report. On the report you’ll get the test data and the ITS-90 calibration coefficients that you can easily input into your Fluke Calibration thermometer.
The 5618B is also a great probe to use for calibrating your Fluke Calibration 9132 or 9133 infrared calibrators. In fact, these IR black body heat sources were designed to be calibrated with this type of probe. Now you can calibrate these targets in your own lab!
For use from –200 °C to 500 °C (the six-inch model goes to 300 °C), you won’t find a better industrial standard in this configuration than our 5618B. We recommend using the 5618B PRTs with the 1523, 1524, 1502A, 1529, or 1560 thermometer readouts.
|Resistance||Nominal 100 Ω at 0 °C|
|Temperature Coefficient||0.003923 Ω/Ω/ °C nominal|
|Temperature Range||–200 °C to 500 °C (–200 °C to 300 °C for 5618B-6-X)|
|Drift Rate||± 0.1 °C when used periodically to 500 °C|
|Sheath Material||316 SST|
|Leads||22 AWG PTFE, 6’|
|Hysteresis||Less than 0.01 °C at 0 °C when using –196 °C and 420 °C as the end points.|
|Time Constant||Four seconds maximum for 63.2 % response to step change in water moving at 3 fps.|
|Thermal EMF||Less than 25 mV at 420 °C|
|Calibration||Includes manufacturer’s NVLAP-accredited calibration w/ITS-90 coefficients, R vs. T values in 1 °C increments, lab code 200706-0|
|Size||5618B-12: 305 mm L x 3.2 mm dia. (12 x 1/8 in) 5618B-9: 229 mm L x 3.2 mm dia. (9 x 1/8 in) 5618B-6: 152 mm L x 3.2 mm dia. (6 x 1/8 in)|
|Calibrated Accuracy† (k=2)||± 0.026 °C at –200 °C ± 0.066 °C at 0 °C ± 0.219 °C at 500 °C|
|†Includes calibration uncertainty and drift specification.||
|2601||Probe Carrying Case, Plastic (for probes up to 305 mm [12 in] in length)|